Calibration tool: pyFAI-calib
=============================

Purpose
-------



Calibrate the diffraction setup geometry based on Debye-Sherrer rings
images without a priori knowledge of your setup. You will need to
provide a calibrant or a "d-spacing" file containing the spacing of
Miller plans in Angstrom (in decreasing order).

If you are using a standard calibrant, look at
https://github.com/silx-kit/pyFAI/tree/master/calibration
or search in the American Mineralogist database:
[AMD]_ or in the [COD]_.
The --calibrant option is mandatory !

Calibrants available:
Al, LaB6, TiO2, Pt, Ni, CuO, quartz, Si, mock, Si_SRM640e, LaB6_SRM660a,
PBBA, cristobaltite, Si_SRM640, NaCl, AgBh, CrOx, LaB6_SRM660c, C14H30O,
Si_SRM640a, Au, alpha_Al2O3, ZnO, Si_SRM640d, Cr2O3, Si_SRM640c,
LaB6_SRM660b, Si_SRM640b, hydrocerussite, CeO2 or search in the American
Mineralogist database: http://rruff.geo.arizona.edu/AMS/amcsd.php

You will need in addition:
 * The radiation energy (in keV) or its wavelength (in A)
 * The description of the detector:

  - it name or
  - it's pixel size or
  - the spline file describing its distortion or
  - the NeXus file describing the distortion

Many option are available among those:
 * dark-current / flat field corrections
 * Masking of bad regions
 * reconstruction of missing region (module based detectors), see option -r
 * Polarization correction
 * Automatic desaturation (time consuming!)
 * Intensity weighted least-squares refinements

The output of this program is a "PONI" file containing the detector
description and the 6 refined parameters (distance, center, rotation) and
wavelength. An 1D and 2D diffraction patterns are also produced. (.dat and
.azim files)

Usage:
------
pyFAI-calib [options] **-w** 1 **-D** detector **-c** calibrant.D
imagefile.edf

options:
--------

**-h**, **--help**
   show this help message and exit

**-V**, **--version**
   show program's version number and exit

**-o** FILE, **--out** FILE
   Filename where processed image is saved

**-v**, **--verbose**
   switch to debug/verbose mode

**-c** FILE, **--calibrant** FILE
   Calibrant name or file containing d-spacing of the reference sample
   (MANDATORY, case sensitive !)

**-w** WAVELENGTH, **--wavelength** WAVELENGTH
   wavelength of the X-Ray beam in Angstrom. Mandatory

**-e** ENERGY, **--energy** ENERGY
   energy of the X-Ray beam in keV (hc=12.398419843320026keV.A).

**-P** POLARIZATION_FACTOR, **--polarization** POLARIZATION_FACTOR
   polarization factor, from **-1** (vertical) to +1 (horizontal),
   default is None (no correction), synchrotrons are around 0.95

**-i** FILE, **--poni** FILE
   file containing the diffraction parameter (poni-file). MANDATORY for
   pyFAI-recalib!

**-b** BACKGROUND, **--background** BACKGROUND
   Automatic background subtraction if no value are provided

**-d** DARK, **--dark** DARK
   list of comma separated dark images to average and subtract

**-f** FLAT, **--flat** FLAT
   list of comma separated flat images to average and divide

**-s** SPLINE, **--spline** SPLINE
   spline file describing the detector distortion

**-D** DETECTOR_NAME, **--detector** DETECTOR_NAME
   Detector name (instead of pixel size+spline)

**-m** MASK, **--mask** MASK
   file containing the mask (for image reconstruction)

**-n** NPT, **--pt** NPT
   file with datapoints saved. Default: basename.npt

**--filter** FILTER
   select the filter, either mean(default), max or median

**-l** DISTANCE, **--distance** DISTANCE
   sample-detector distance in millimeter. Default: 100mm

**--dist** DIST
   sample-detector distance in meter. Default: 0.1m

**--poni1** PONI1
   poni1 coordinate in meter. Default: center of detector

**--poni2** PONI2
   poni2 coordinate in meter. Default: center of detector

**--rot1** ROT1
   rot1 in radians. default: 0

**--rot2** ROT2
   rot2 in radians. default: 0

**--rot3** ROT3
   rot3 in radians. default: 0

**--fix-dist**
   fix the distance parameter

**--free-dist**
   free the distance parameter. Default: Activated

**--fix-poni1**
   fix the poni1 parameter

**--free-poni1**
   free the poni1 parameter. Default: Activated

**--fix-poni2**
   fix the poni2 parameter

**--free-poni2**
   free the poni2 parameter. Default: Activated

**--fix-rot1**
   fix the rot1 parameter

**--free-rot1**
   free the rot1 parameter. Default: Activated

**--fix-rot2**
   fix the rot2 parameter

**--free-rot2**
   free the rot2 parameter. Default: Activated

**--fix-rot3**
   fix the rot3 parameter

**--free-rot3**
   free the rot3 parameter. Default: Activated

**--fix-wavelength**
   fix the wavelength parameter. Default: Activated

**--free-wavelength**
   free the wavelength parameter. Default: Deactivated

**--tilt**
   Allow initially detector tilt to be refined (rot1, rot2, rot3).
   Default: Activated

**--no-tilt**
   Deactivated tilt refinement and set all rotation to 0

**--saturation** SATURATION
   consider all pixel>max*(1-saturation) as saturated and reconstruct
   them, default: 0 (deactivated)

**--weighted**
   weight fit by intensity, by default not.

**--npt** NPT_1D
   Number of point in 1D integrated pattern, Default: 1024

**--npt-azim** NPT_2D_AZIM
   Number of azimuthal sectors in 2D integrated images. Default: 360

**--npt-rad** NPT_2D_RAD
   Number of radial bins in 2D integrated images. Default: 400

**--unit** UNIT
   Valid units for radial range: 2th_deg, 2th_rad, q_nm^-1, q_A^-1,
   r_mm. Default: 2th_deg

**--no-gui**
   force the program to run without a Graphical interface

**--no-interactive**
   force the program to run and exit without prompting for refinements

**-r**, **--reconstruct**
   Reconstruct image where data are masked or <0 (for Pilatus detectors
   or detectors with modules)

**-g** GAUSSIAN, **--gaussian** GAUSSIAN
   Size of the gaussian kernel. Size of the gap (in pixels) between two
   consecutive rings, by default 100 Increase the value if the arc is
   not complete; decrease the value if arcs are mixed together.

**--square**
   Use square kernel shape for neighbor search instead of diamond shape

**-p** PIXEL, **--pixel** PIXEL
   size of the pixel in micron


Tips & Tricks
-------------

The output of this program is a "PONI" file containing the detector
description and the 6 refined parameters (distance, center, rotation)
and wavelength. An 1D and 2D diffraction patterns are also produced.
(.dat and .azim files)


Example of usage:
-----------------

Pilatus 1M image of Silver Behenate taken at ESRF-BM26:
.......................................................

::

	pyFAI-calib -D Pilatus1M -c AgBh -r -w 1.0 test/testimages/Pilatus1M.edf

We use the parameter -r to reconstruct the missing part between the modules of the
Pilatus detector.


Half a FReLoN CCD image of Lantanide hexaboride taken at ESRF-ID11:
...................................................................

::

	pyFAI-calib -s test/testimages/halfccd.spline -c LaB6 -w 0.3 test/testimages/halfccd.edf -g 250


This image is rather spotty. We need to blur a lot to get the continuity of the rings.
This is achieved by the -g parameter.
While the sample is well diffracting and well known, the wavelength has been guessed.
One should refine the wavelength when the peaks extracted are correct


All those images are part of the benchmark suite of pyFAI.
To download them from internet, run the benchmark:

::

	pyFAI-benchmark

Downloaded test images  are located in /tmp/pyFAI_testdata_${USER}
